This instruction manual provides information specific to the AIRANGER XPL PLUS (Ten Point Level) monitor. When references are made to other Milltronics. AiRanger XPL Plus. * Program range is defined as the empty distance to the face of the transducer (P) plus any range extension (P). **EMC performance. AiRanger XPL Plus. Ten-Point Ultrasonic Level Measurement System. Part 1. General. Scope. A. This section describes the requirements for an ultrasonic.
|Published (Last):||16 February 2005|
|PDF File Size:||6.3 Mb|
|ePub File Size:||18.1 Mb|
|Price:||Free* [*Free Regsitration Required]|
Milltronics AiRanger XPL Plus Instruction Manual
Page Extend Near Blanking P to a distance from the transducer face to just beyond the distance represented by the Reading.
Level or Material Volume Measurement for a single vessel. Page 44 Allocation P range. Add to basket. Page 4 Measurement Parameters P to P By pressing ‘print’ button you will print only current page. Select “short” to have only short shots fired if the Echo Confidence P produced by a short shot exceeds the short Confidence Threshold P Extend Near Blanking P to a distance from the transducer face to just beyond the distance represented by the Reading.
Installation Application Examples for additional installation requirements. A positive value is entered for a filling rate alarm, negative for emptying.
Sign in for checkout Check out as guest. Normal Operation indicates operating conditions are good and the Reading is reliable. Don’t show me this message again.
To copy a parameter set with the desired transducer point selected: The seller won’t accept returns for this item. Enter the delay in seconds before the display advances to the next Point Number.
Milltronics airanger XPL plus Instruction Manual
No response to programmer. Redeem your points Conditions for uk nectar points – opens in a new window or tab.
From this graph, determine which Universal Volume calculation will provide the best results, and select the level vs volume breakpoints to be entered 32 max. Table of Contents Airaanger to my manuals Add. This register increments by 1 each time a bus error P is reported.
Press to scroll access the failsafe options. If Empty is substantially higher than the actual vessel bottom, increase Range Extension such that Empty plus Range Extension is greater than the transducer to vessel bottom distance.
Relays may be programmed as Standard Alarms or for Custom Relay operation. Page 68 The noise level is a combination of transient acoustic noise and electrical noise induced into the transducer cable or receiving circuitry itself.
Enter the point number to copy from. Wipe out the area of the enclosure lid docking bay recess with a clean dry cloth if necessary before installing the programmer. In plua cases, when a Parameter Value is altered, associated Parameter Values are automatically altered accordingly.
MILLTRONICS AIRANGER XPL PLUS INSTRUCTION MANUAL Pdf Download.
If “FAIL” is displayed, repeat the test. P Span Enter “3” for 3 ft vessel diameter. If the material presents a flat surface perpendicular to the transducer beam, select liquid. Based on level, space, distance, or volume, Watch list is full. If the material monitored is a liquid, check xll splashing in the vessel. Add to Watch list Watching Watch list is full.